Strap Auxiliary Equipment
Tensile testing machine
1. Equipment usage
Microcomputer control electronic universal tester can operate mechanics performance testing and analyst on metallic-materials, non-metallic materials. Tests can be carried out on the stretching, compression and bending of the material according to GB, ISO, ASTM, JIS, DIN, JC/JCJ and some other standard. The tensile strength, elongation at break, elastic at break, elastic modulus and other parameters of the material can be detected.
2. Test and control system introduction
Microcomputer control electronic universal tester adopt microcomputer control and combined with advanced electronic control technology, standardized unitized design. It is characterized by accurate control, high measurement accuracy, flexible configuration, easy accessory exchange and easy after-sales service.
3. Technical specifications of the tester
3.1. Force transducer is the wheel type tension and pressure transducer. Precision 0.05%FS, force measuring system precision is in the range of 2%~100%. Relative accuracy is up to 0.01mm.
3.2. Electronic extensometer, precision grand is C.
3.3.displacement transducer adopt 600 rings optical-electricity encoder. The resolution can be up to 0.01mm.
4. Performance feature
4.1.Batch test: a batch of samples can be tested one time after setting the same parameters.
4.2.Test report: (standard programs)
4.2.1.stretch: Max strength, tensile strength, elongation at break, modulus of elasticity.
4.2.2.Bending: Max strength, bending strength, deflection, modulus of elasticity.
4.2.3.Compression: Max strength, compression strength, yield strength, modulus of elasticity. Test reports can also be programmed and printed in a format required by the user.
5. technology parameters
5.1.Max test force: 20kN
5.2.Measuring range of test force: 0-20kN
5.3.Accuracy of test force: is superior to ±1% of the indicated value.(0.4kN-20kN range)
5.4.Resolution of test force
5.5.Speed accuracy is superior to ±1% of the indicated value.